[time-nuts] GPSDO Question
w1ksz at earthlink.net
Fri Aug 31 17:52:54 EDT 2007
If you use the HP Oscillator in your GPSDO, you can then use the
10 MHz output as an external reference for your counter.
I picked up a 10 MHz distribution amp and use the same GPSDO to
act as external reference on several instruments.
Regards, Dick, W1KSZ
Tom Clifton wrote:
> ); SAEximRunCond expanded to false
> Errors-To: time-nuts-bounces+w1ksz=earthlink.net at febo.com RETRY
> I have gathered everything I need to assemble a GPS
> Disciplined Oscillator. I have the Brooks Shera
> controller assembled, am about 99% done on the linear
> power supplies for +/- 5V, 12V supplies and a separate
> 24V oven heater supply. The GPS Engine is purring away
> happily with a good active antenna - spewing NMEA
> data and 1pps to anything willing to listen to it.
> The HP-10811 oven is sitting happily in a working
> 5328A counter that I'm willing to cannibalize, but do
> I need to? Is there any reason why I shouldn't put the
> controler in the 5328 case so I can use it as a
> counter as well as timing source?
> 73 - Tom in St. Louis
> Got a little couch potato?
> Check out fun summer activities for kids.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
73, Dick, W1KSZ
More information about the time-nuts