[time-nuts] Thunderbolt Power Supply Question
golgarfrincham at yahoo.com
Mon Apr 26 22:03:04 UTC 2010
Ed Palmer-"Every oscillator has a spec for frequency shift due to power supply variation,
but maybe you'd see a big change when you lost regulation. Have to think about that."
I have done this a few times with oscillators that contain internal regulators and the output
level remains constant and the frequency display on my scope (triggered by a standard)
remains stable until you reach the point where the internal regulator output starts to drop
and you do see a significant change. Because the oscillator and buffer stages generally
only draw a few ma., a low power linear regulator with good filtering is easy to design.
HP says this about the simple circuit they used in the 10811:
"8-25. VOLTAGE REFERENCES (5.7V AND 6.4V)
8-26. Constant current diode CR2 feeds 1 mA to zener diode CR3 providing 6.4Vdc for the
EFC varactor reference. R12 and C15 form a filter to attenuate noise from the zener
As to the -12VDC supply, one unit that used the Thunderbolt used -7VDC instead of
-12VDC and the same Thunderbolt works just fine with either voltage.
More information about the time-nuts