[time-nuts] ADEV vs MDEV
magnus at rubidium.dyndns.org
Sat Feb 6 13:31:56 UTC 2010
Bruce Griffiths wrote:
> Oops, Its the CNT91 not the CNT81 that actually does the regression fit.
> How did you achieve MDEV ~1E-13 @10s with a counter rated at 50ps single
> shot resolution?
I think the actual resolution and trigger jitter was combined to a
single number. For some reason they do that. You need to look more into
the details of the counters.
More information about the time-nuts