[time-nuts] FE-5680A's suitability for use as a 10 MHz reference for microwave transverters
bruce.griffiths at xtra.co.nz
Tue Feb 14 02:09:30 UTC 2012
paul swed wrote:
> Clint a good read and a fine approach.
If and only if the high phase noise pedestal exhibited by the buffer
amplifier that extends to offsets of a few hundred kHz or so isn't an issue.
The relatively low isolation between the 10MHz outputs may also be an issue.
> By accident I looked at the original filter schematic first and believe it
> has an error for the first output amplifier. The 470 ohm resistors in the
> wrong location.
> Others on the list have mentioned the same thing about the 5680s output
> being dirty. You have that fixed up nicely.
> On Mon, Feb 13, 2012 at 4:03 PM, C. Turner<turner at ussc.com> wrote:
More information about the time-nuts