[volt-nuts] Tilt effect on Keithley 2000
Andreas_-_Jahn at t-online.de
Sun Sep 5 07:21:14 UTC 2010
>What is the time constant? Slowish (i.e., could be thermal) or rapid
>(presumably gravitational/mechanical stress)?
The 20-30uV on my LM399s should be thermal. Time constant
is about 10 seconds. Additionally the current consumption of
the reference cirquit changes by 3-5%. So I think that the
heater within LM399 has not a good thermal layout with respect
to the temperature sensor on the same chip.
For the Keithley differences I cannot tell the real time constant
since I had slow reading with averaging 10 or 50 measurements.
But since the effect is significantly larger than the reference alone
there might be a mixture of mechanical and thermal effects.
More information about the volt-nuts