[time-nuts] Cesium DOES Drift, not
brooke at pacific.net
Wed Apr 20 12:32:34 EDT 2005
It's just that my FTS4060 IS drifting.
The Excel equation and fit for a month of data as of this morning are:
y = -1.2954x2 + 243.1x - 10631
R2 = 0.9607
Where y is in ns, x is the Day of Year and R2 is the quality of the fit
The first derivative (after dividing by 86400) gives:
Offset = -3E-14 * X + 243
i.e. minus 3E-14 per day.
I'll keep watching and post from time to time, maybe this will stabilize???
Brooke Clarke, N6GCE
w/o Java http://www.pacificsites.com/~brooke/PRC68COM.shtml
Richard (Rick) Karlquist (N6RK) wrote:
>My understanding from talking to the experts is
>that no aging, or environmental effects are observable
>on the 5071A, and measurements have been made into
>the 10^-15 range. Also, no systematic frequency offset
>has been observed down to at least 1x10^-14. IE,
>frequency offset between different 5071A's is randomly
>distributed with a mean within 10-14 of laboratory
>standards, such as Cs fountains. Knowing the way we
>overdesigned the 5071A, I am not surprised that it
>is this good.
>(member of 5071A design team)
>time-nuts mailing list
>time-nuts at febo.com
More information about the time-nuts