[time-nuts] FTS4060 C Field Time Constant
brooke at pacific.net
Thu Feb 17 16:18:48 EST 2005
In still in the process of setting the C field on my FTS-4060 and have an interesting observation. After a cockpit error where I made a change in the wrong direction the time interval plot was going up steeply. In line with my latest theory of how to make the adjustment (which is to do a binary search rather than try and compute the required adjustment) I then set in a new value. I thought I saw a time constant of about 3 hours (i.e. it took about 3 hours to get to 63% of what I thought was the final value).
The next day (24 hours later) for about 8 hours the time interval was constant to within 10 nano seconds, phenomenally stable operation.
Now 3 days after that I can see that the time interval has a down slope. So the very stable operation 24 hours after the C field change was during the time when the frequency was changing and was at the level part of the curve.
This indicates that for a C field change on the FTS4060 you need to wait at least 2 or 3 days before making any measurements!
This also may explain why trying to compute the amount of a change does not seem to work. This would be the case if the effect of the prior change had not yet settled in.
Brooke Clarke, N6GCE
w/o Java http://www.pacificsites.com/~brooke/PRC68COM.shtml
More information about the time-nuts