[time-nuts] Failure mode in GPS receivers
dwsmith at bigpond.net.au
Sun Dec 17 18:25:49 EST 2006
I've owned a Trak Systems 8821 for over a year, and had a problem with
the antenna side. Yours may be different.
My unit is designed to work with a powered antenna, and it feeds 12V up
the coax to power it. Apparently this is the "standard" voltage for TNC
antenna connections - the smaller MCX etc. are 5V standard.
In my ignorance, I initially connected one of the regular "mouse" type
antennas to it that requires 5V. The antenna worked OK for months, and
I only discovered the error when I was fitting another more permanent
antenna. Strangely, the "mouse" no longer works on 5V, but seems to get
by on 12V.
Because most powered antennas these days seem to be 5V, I modded the
Trak with a 78L05 in the antenna power circuit to match.
So, what you could have done is fried the port on your splitter with the
12V? Just a guess.
P.S. Have you got any software for yours? I've got a User Manual PDF
for mine if that helps.
More information about the time-nuts