had at to-way.com
Sun Feb 19 19:03:28 EST 2006
More than likely the internal back up battery is dead and it lost
it's stored cal. Hopefully the battery has not leaked and eaten up any traces.
At 03:10 PM 2/19/2006, you wrote:
>I just acquired a HP5372A Frequency and Time Interval Analyzer. When I
>power it up I get the following error:
>Error 160: Out of sensitivity cal.
>I am currently downloading the manuals but I'm sure that someone on
>time-nuts has seen this error and know what it takes to fix this.
>Any ideas folks?
>time-nuts mailing list
>time-nuts at febo.com
A well regulated Militia, being necessary to the security of a free
State, the right of the people to keep and bear Arms, shall not be
More information about the time-nuts