[time-nuts] Some results of PRS10 and Trimble Resolution T
ejkroon at hetnet.nl
Mon Jun 26 14:14:14 EDT 2006
At this moment I have only a interval of 6 sec data for the PRS10.
Now a am measuring the output of the Trimble GPS.
After the measurement I will do a new PRS10 measurement using small
I use the prefilter and a PT setting of 5. The powersupply I use is a linear
I measure direct against a HP5071 corrected by a micro phase stepper and
using a SR620.
The HP5071 is one of four clocks we also use for the calculation of UTC at
our laboratory NMi VSL
I did not see your data, it was not in the mail.
----- Original Message -----
From: <SAIDJACK at aol.com>
To: <time-nuts at febo.com>
Sent: Monday, June 26, 2006 7:44 PM
Subject: Re: [time-nuts] Some results of PRS10 and Trimble Resolution T
> Hello Erik,
> I recently measured my PRS10 as well.
> You can see a big hump (about 1.6E-011) at the 2s measurement intervall.
> While this is within spec, does anyone know why the unit is so noisy at
> particular intervall? Is there a filter setting that would improve this?
> Could this be power supply injected noise?
> I have set PT to 7, driving the unit with an Motorola M12+.
> I measured the ADEV with the new TSC5120A. Other oscillators I have
> at around 1E-012 at 2s, so it's not a setup problem.
> time-nuts mailing list
> time-nuts at febo.com
More information about the time-nuts