[time-nuts] Time Interval measure mesurements
blanning at acadia.net
Mon Jun 26 17:11:18 EDT 2006
Hi John, Thanks for the prompt reply. I am Measuring the 10 MHz with the
counter set at .1usec. The display says us. I just turned the setup on again
and now I get gust the count every time the time base opens. It reads
between .0 and .4. I trust this helps you John. Give me your phone # and a
good time to call and I will call you.
----- Original Message -----
From: "John Ackermann N8UR" <jra at febo.com>
To: "Discussion of precise time and frequency measurement"
<time-nuts at febo.com>
Sent: Monday, June 26, 2006 5:00 PM
Subject: Re: [time-nuts] Time Interval measure mesurements
> Bruce Lanning wrote:
>> I am trying to compare my Freq, Std, HP-Z3801A GPS Receiver, to my home
>> Freq Std. using my Freq. counter, a HP-5245L with a Time Interval Unit
>> (HP-5262A) Plug in. Can someone help me. My counter Time Base opens about
>> 6-8 times and then gives me a reading like 1,345,000.1 My problem is the
>> to 8 times that the Time Base opens and then the reading like the one I
>> The manual on the counter and the Time Interval Unit give me no
>> Can one of you Time-Nuts out there tell me what is going on or is my Time
>> Interval Unit defective?
>> Bruce W1GBS
> Bruce, can you give a little more detail about the measurement setup?
> In particular, are you measuring 1pps or 10MHz, and what mode is the
> counter set to? Which signal is hooked up to what input? Is there any
> indication on the display of what the units are?
> time-nuts mailing list
> time-nuts at febo.com
More information about the time-nuts