[time-nuts] ...Your thoughts on the Trimble Resolution T?
Tom Van Baak
tvb at leapsecond.com
Fri Mar 31 08:40:20 EST 2006
You can find M12+ sawtooth information and plots at:
Also, here is a nice graph comparing the 1 PPS error
of 4 GPS receivers:
Dark blue is a Motorola Oncore VP with approximately
+/- 50 ns of jitter; 30 ns rms.
Pink is a Motorola M12+, 10 ns rms.
Yellow is a Motorola M12+ with hardware (or software)
sawtooth correction; 6 ns rms.
Green is an HP SmartClock (58503B) with 1.3 ns rms.
Note however that although the short-term jitter is less
there appears to be more wander of the phase mid-term
(typical of a GPSDO).
So in the end, for long-term 1 PPS timing use, I found
the M12+ with sawtooth correction more stable than the
other GPS receivers, including a GPSDO.
More information about the time-nuts