[time-nuts] Stepping up the output of an OCXO
tractorb at ihug.co.nz
Sat Feb 3 01:10:44 EST 2007
Not sure if the NVE Isoloop range of GMR based high speed data
couplers would have any advantages in this regard-their data is a bit
sketchy but they would likely have similar issues- aside from
certainly being fast enough.
----- Original Message -----
From: "Dr Bruce Griffiths" <bruce.griffiths at xtra.co.nz>
To: "Don Collie" <donmer at woosh.co.nz>; "Discussion of precise time and
frequency measurement" <time-nuts at febo.com>
Sent: Saturday, February 03, 2007 6:43 PM
Subject: Re: [time-nuts] Stepping up the output of an OCXO
> Don Collie wrote:
>> Why not use an optocoupler as an isolation amplifier? - to the best
>> of my
>> knowledge it would provide infinite isolation.
>> Cheers,.......................................Don Collie
> And lots of noise.
> You will need a cleanup PLL on the output side.
> Also optocoupler isolation isn't infinite, there is a finite
> between the closely spaced emitter and detector.
> Not all optoisolators are fast enough.
> The most effective way of using optical isolation is in fact to
> a stabilised laser and then couple the output into a fibre.
> This technique is employed to distribute stable frequencies to Radio
> telescope antennas especially when a large number are employed in an
> A phase locked loo is used at the other end to cleanup the signal
> the photodetector.
> Fibre has the advantage in this and other applications where long
> are involved that its propagation delay tempco is significantly
> than that of coax.
> It is also possible to use fibre stretchers and other techniques to
> actively compensate variations in the fibre propagation delay.
> Isolation becomes particularly important when one mixes a pair of
> signals a few Hz apart and then analyses the zero crossing times of
> mixer output beat signal.
> It is easier to achieve a high isolation between RF frequencies than
> frequencies of a few Hz, JPL found fibre optic isolation can be
> invaluable in this case.
> time-nuts mailing list
> time-nuts at febo.com
> No virus found in this incoming message.
> Checked by AVG Free Edition.
> Version: 7.5.432 / Virus Database: 268.17.19/663 - Release Date:
> 1/02/2007 14:28
More information about the time-nuts