[time-nuts] HP 5370A Frequency counter Problem(s)
jmiles at pop.net
Wed Mar 21 21:50:05 EDT 2007
I just pull the board (it's fairly obvious how, but I believe the manual
also talks about how to disassemble the front panel), take the sliders out
of the switches, and swab everything down with Caig ProGold.
The only tricky part is paying attention to how the springs are installed on
the switch sliders. If you put them back the wrong way, you can bend or
break the spring when you try to operate the switch.
-- john, KE5FX
> -----Original Message-----
> From: time-nuts-bounces at febo.com [mailto:time-nuts-bounces at febo.com]On
> Behalf Of jshank
> Sent: Wednesday, March 21, 2007 6:30 PM
> To: Discussion of precise time and frequency measurement
> Subject: Re: [time-nuts] HP 5370A Frequency counter Problem(s)
> Could someone please provide me with a link or the procedure to
> cleaned / lubed.
More information about the time-nuts