[time-nuts] 5370B frequency counter capability
John Ackermann N8UR
jra at febo.com
Tue Sep 25 08:50:38 EDT 2007
I was looking at the wrong plot when I mentioned the noise floor of my
5370B in frequency counter mode; blame my temporary mono-vision after
having had some eye surgery!
If you look at http://www.febo.com/time-freq/hardware/5370B/index.html
*properly*, you'll see that the actual noise floor is about 3.7x10e-10
in 1 second, roughly an order of magnitude worse than the time interval
noise floor. I also noticed that I ran that test five years ago... I
should try it again with some of my current equipment.
Sorry about that...
More information about the time-nuts