[time-nuts] Generation of very stable 10.001MHz for heterodyne freq measurement
Richard (Rick) Karlquist
richard at karlquist.com
Fri Jan 25 14:38:06 EST 2008
Or look at the 1995 FCS proceedings.
John Ackermann N8UR wrote:
> Anders Time said the following on 01/25/2008 07:23 AM:
>> What is the best way to generate a very stable 10.001MHz(low E-13 or
>> -110dBc/1Hz) to be to do high resolution heterodyne(Allan deviation)
>> Using DDS?
>> Using PTS 500 or 250 synth and divide? Using HP 8662A and divide?
>> Some smart pll circuit by locking the 1000Hz to the reference in some way?
> Rick Karlquist has published a design for a very low noise synthesizer
> designed for just this application. Unfortunately, I don't have a link
> to that handy, but it's on his website.
> I just took a look at some measurements I did on a PTS-250-SX51 (special
> version with low noise output to 25 MHz via an internal divider) and it
> looks like it would barely meet your requirements -- around 2x10e-13 at
> 1 second, and about -115 dBc/Hz at 1 Hz offset:
> However, that was at 10.000... MHz; the performance, particularly with
> respect to spurs, changes as the decimal dividers kick in for more
> complex frequencies.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts