[time-nuts] Zeta Labs X76 Multiplier 7600 MHz Output Model 5856-01
brooke at pacific.net
Tue May 6 14:13:46 EDT 2008
Is it possible to make a DC measurement (DMM Diode function) of the SRD forward
and reverse voltage?
http://www.prc68.com/P/Prod.html Products I make and sell
http://www.prc68.com/Alpha.shtml All my web pages listed based on html name
http://www.prc68.com/I/WebCam2.shtml 24/7 Sky-Weather-Astronomy Web Cam
Matt Ettus wrote:
> On Mon, Apr 28, 2008 at 4:15 PM, Rick Karlquist <richard at karlquist.com> wrote:
>>I worked for Zeta Labs many years ago. This multiplier
>> undoubtedly has a step recovery diode that is being driven
>> at a level of a good watt or two. In most cases there was
>> a hefty transistor to make this power. You probably need
>> +10 dBm or so to drive it, maybe as much as +20 dBm. Take
>> the cover off and look at the input section and it should
>> be apparent if there is any additional gain. You can always
>> start at 0 dBm and work up, looking at the power out as you
>> go along. The adjustment of these can be very tricky, so
>> proceed with extreme caution.
> I just got the multiplier. It draws about 220mA at 15V, but won't
> produce any output. I've given it a wide range of input powers from
> very low up to about 20dBm. I took off the bottom cover, but it only
> exposes the back of a very complex circuit board. The only accessible
> parts are the trimmer caps which come through the PCB.
> Any suggestions?
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts