[time-nuts] HP 5370B
didier at cox.net
Sat May 10 10:43:01 EDT 2008
I knew would get comments on that. I simply did not provide enough
information. I thought most everyone who has done it would know what I was
Acuracy of these instruments is not one number, it is a 3 dimensional chart,
and time is the 4th dimension.
Forget I even mentioned it. I do not intend to go into an error analysis
> -----Original Message-----
> From: time-nuts-bounces at febo.com
> [mailto:time-nuts-bounces at febo.com] On Behalf Of Mike S
> Sent: Saturday, May 10, 2008 9:19 AM
> To: Discussion of precise time and frequency measurement
> Subject: Re: [time-nuts] HP 5370B
> At 09:22 AM 5/10/2008, Didier Juges wrote...
> >Most microwave network analyzers have amplitude resolution
> of 0.01dB,
> >while their accuracy is just around 1dB in most cases.
> >I have had to argue too many times that a piece of equipment
> with a 2dB
> >p-p requirement on flatness was just fine when it measured 2.01dB on
> >the HP network analyzer. I would not have gotten in that argument if
> >the data had been 1.99dB. Go figure.
> I figure you're both wrong. If the measuring instrument is no
> better than 1 dB absolute, you can't expect to do it - a flat
> line measurement could represent an actual + or - 1 db, the
> limit of the spec.
No virus found in this outgoing message.
Checked by AVG.
Version: 7.5.519 / Virus Database: 269.23.11/1422 - Release Date: 5/8/2008
More information about the time-nuts