[time-nuts] 5061B info.
corby d dawson
cdelect at juno.com
Sun Oct 5 16:46:03 UTC 2008
What is the 2nd harmonic meter reading?
If it is too low you will not be able to get the operate light on.
While watching the 2nd harmonic reading adjust the front panel beam I pot
for a reading of 35 to 45.
If pot is at max adjust the loop gain thru the front panel hole for the
35-45 reading, if it's at max and the top switch on the A7 is in low
gain, switch it to high and try the loop gin adjust again.
If the 2nd harmonic is OK but still no operate light, check the operate
lamp, if OK then remove the bottom cover and find the two wires pushed
onto the connections on the synthesizer board. Leave the red one
connected but remove the other.
Now push the logic reset and see if the operate light comes on.
If so then the sythesizer is not properly locked and is your problem.
All is not lost! Click now for professional data recovery.
More information about the time-nuts