[time-nuts] What is the best counter for a Time Nuts?
Tom Van Baak
tvb at LeapSecond.com
Thu Oct 9 14:54:56 UTC 2008
> Hi Tom, Ulrich,
> can't seem to get 12 digits/s out of my 53132A in frequency mode when doing
> measurement offset of -10MHz. I only get 11 digits/s for some reason. Have to
> go to 10s + gate time to get better resolution.
Correct. When you get "close" to an exact 10 MHz input then
the counter drops down into a "reduced resolution" mode and
you lose a digit (regardless of gate time). If you need, you can
query this with the :DIAG:MEAS:PRES? command. Details are
in the back of the 53132A manual.
More information about the time-nuts