[time-nuts] time-nuts Frequency Divider
bruce.griffiths at xtra.co.nz
Thu Apr 2 23:28:31 UTC 2009
The problem is more accurately described as:
When the bias network dc level at the 74AC04 (or 74HC04) inverter input
isn't equal to the switching threshold of the particular device then AM
modulation on the input signal is converted to phase noise as switching
no longer occurs at the zero crossing of the input signal.
Such behaviour is inherent when using a Schmitt trigger circuit and it
cannot be cured with a feedback circuit that stabilises the output duty
A well designed limiter + filter cascade in front of the comparator,
Schmitt trigger or logic gate can be used to minimise such AM to PM
conversion whilst minimising the output jitter.
Mike Monett wrote:
> > Message: 3
> > Date: Fri, 03 Apr 2009 09:04:59 +1300
> > From: Bruce Griffiths <bruce.griffiths at xtra.co.nz>
> > Subject: Re: [time-nuts] Frequency Divider
> > Hal Murray wrote:
> >>> A large resistor connected between the input and output would
> >>> accommodate threshold variations better. Even better would be a
> >>> feedback loop that adjusts the input bias point to maintain the
> >>> output duty cycle at 50%.
> >> Isn't that resistor a feedback loop?
> >> I played with that setup in the lab many years ago. It didn't
> >> work as well as I was expecting. I didn't figure out why it
> >> didn't work better.
> >> Maybe some gain in the feedback path would help. Then we have to
> >> consider stability. Ugh.
> > Hal
> > Yes, a resistor connected between the input and output of an
> > inverter is a feedback loop but the loop gain is relatively low.
> > With a high amplitude input threshold variations from the nominal
> > can cause the input protection diodes to conduct.
> > Once these diodes conduct the output jitter may deteriorate
> > significantly (it does for HCMOS inverters).
> > Using a non inverting integrator in the feedback path can
> > accurately stabilise the duty cycle.
> > Bruce
> The 74HC and 74AC input threshold tolerance is +/- 30%. This means
> the threshold can vary from 1.5V to 3.5V with a Vcc of 5V.
> This limits the maximum input signal to 3V p-p or +13.5dBm, and
> leads to a very subtle flaw discovered in some amazing engineering
> work by Martein Bakker, PA3AKE.
> If the threshold is not controlled, it can cause AM noise to convert
> to PM noise and degrade the jitter. This occurs in the Analog
> Devices AD9910 1GHz DDS chip.
> Martein Bakker discovered this in his noise analysis, and Kevin
> Wheatly gave a nice entry in his blog on how to fix it:
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts