[time-nuts] Tbolt temperature sensor
holrum at hotmail.com
Thu Feb 5 00:03:29 UTC 2009
All my earlier Tbolts report 2002-2003 manufacturing date codes. They are rev 2.22 and 3.00 firmware. They all show the nice smooth high res temperature curves that are usually sinusoidal. I know of around a dozen units that respond this way.
All the later build Tbolts that I have seen (mine, Asa's, and one other) are from the March to May 2005 time frame. They all show the basically flat temperature curve that is quantized to 0.25/0.75C readings (except when a firmware filter kicks in for a short time when the value steps between readings).
The temperature sensor chip is a DS1620 chip in a small SOIC-8 package next to the RS-232 connector. I guess the questions are:
1) Is the difference a firmware change (even though the "problem" units all report the same 3.0 firmware levels as the earlier units) or is it a change in the sensor chip that causes the high-res temperature read technique listed in the DS1620 data sheet to fail on the Tbolts?
2) Which units are affected by the issue? Can it be isolated to a date code or some other marking on the DS1620 package? Is there a Tbolt manufacturing date or range where the issue first pops up or occurs over?
Lady Heather shows the Tbolt manufacturing dates and firmware dates. She also plots the temperature curve.
Windows Live™: Keep your life in sync.
More information about the time-nuts