[time-nuts] About HP10544A
bruce.griffiths at xtra.co.nz
Sat Mar 7 21:36:49 UTC 2009
Esa Heikkinen wrote:
>> New 10544A's have a somewhat cleaner output than that.
>> Perhaps something is wrong with the emitter follower.
> Sorry I forgot to say that the oscilloscope picture was taken directly
> from HP10544A output, without emitter follower but with 1:1 probe,
> output terminated with 1 kohm. resistor. With spectrum analyzer the
> emitter follower is necessary due to 50 ohm. input impedance of the
> Datasheet of HP10544A claims that harmonics should be more than 25 dB
> down. Doesn't look like they are.
> Here's a spectrum measured from OCXO which is used in the Thunderbolt.
> I'm not an crystal oscillator guru but I think that this is just what an
> xtal oscillator output should be:
Please provide a circuit schematic for your emitter follower buffer.
Include transistor types and component values as well as power supply
More information about the time-nuts