[time-nuts] OT: leaching was, Alternative time interval interpolation technique
bruce.griffiths at xtra.co.nz
Sun Aug 15 19:58:01 UTC 2010
Attila Kinali wrote:
> On Sun, 15 Aug 2010 14:59:43 +1200
> Bruce Griffiths<bruce.griffiths at xtra.co.nz> wrote:
>> Mea Culpa.
>> Below is a link to the paper using SAW filters to achieve a sub ps time
>> interval interpolator noise:
>> And the associated presentation:
> A quick skimming over the error analysis Panek made, suggests that
> the jitter of the clock source is the biggest contributor to measurement
> errors. But he never says how a clock source with such a low jitter is
> build. Although he references a few times a module build by Josef Kölbl
> of the Fachhochschule Deggendorf, there is no description available what
> kind of device that is.
> Does anyone have any pointers to recommended reading on the design of such
> low jitter oscillators?
> Attila Kinali
Such low jitter oscillators are readily available.
With some care (bandpass filtering) a cycle to cycle jitter of around
50fs or so is attainable with a Wenzel OCXO for example.
However the time interval jitter degrades as the time interval increases.
Achieving a cycle to cycle jitter of 1ps or so is relatively easy with a
10MHz or 100MHz OCXO having sufficiently low phase noise.
More information about the time-nuts