[time-nuts] Austron 2100F
stanw1le at verizon.net
Fri Oct 22 00:41:35 UTC 2010
Do you have the manual ?
My june, 1983 2100F op and maintenance manual, page 3-13 indicates:
E #1 is a - or acquisition not in progress, cleared by software
E#2 is a E, normal tracking has been halted due to a loss of external
The operator clears this bit manually by restoring the external
E#3 is a - , noise number is less than 4000
E#4 is a - , no blinking
E#5 is a - , servos active
E#6 is a E not yet in tracking mode . this bit is cleared by software.
On page 3-14: The STATUS indicators are displayed by (keystrokes) STATUS 6
To clear the indicators, enter 0 STATUS 6
I found no other method of a deep reset.
Take a look at the external reference connection to be sure it is
Any questions, send them along.
Stan, W1LE Cape Cod
On 10/21/2010 6:55 PM, Chuck Harris wrote:
> Are you sure that your Loran chain is still up? In the US and
> Canada, Loran has been decommissioned.
> -Chuck Harris
> asmagal at fc.up.pt wrote:
>> After power up, my 2100F Loran-C receiver displays
>> (E 1002048) followed by (E-20), as usually, but recently
>> I have a trouble: after that the TRACKING led goes RED
>> still and the display shows (- E - - - E).
>> Running the test subroutines 1 to 4 everything seems
>> to be OK but the fault persists.
>> Entering the sequence (0, STATUS, 6) doesn't clear the
>> Does someone knows how to make a deep reset for
>> clearing all the errors?
>> Thanks in advance.
>> time-nuts mailing list -- time-nuts at febo.com
>> To unsubscribe, go to
>> and follow the instructions there.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the time-nuts