[time-nuts] What is the correct Allan value?
mike.cook at orange.fr
Fri Oct 29 19:57:12 UTC 2010
looks like this might run...
I would say that all the values are valid , but agree that the overall
confidence in the performance couldn't be better than the worst case. I
have noticed that I never get the same ADEV values on successive runs
but put it down to an undisciplined thermal environment. I suppose that
if you were running an experiment you would ideally measure the ADEV
over the periods of interest and use those values as opposed to relying
on some longer baseline.
Le 29/10/2010 21:20, Poul-Henning Kamp a écrit :
> In message<962BA941C5CA41C19DF45F77608AD38F at Warcon28Gz>, "WarrenS" writes:
>> What is the correct 100 sec Allan value of this low noise 10811 Osc?
>> All plots are of the same OSC, just different parts of the same data run.
> Basically the worst one...
More information about the time-nuts