[time-nuts] Thunderbolt oven / non-stable operating temperature
stanw1le at verizon.net
Tue Dec 11 02:01:51 UTC 2012
On occasion I get an "ALARM 4000" with Lady Heather V3.10,
However I know not what it means. The alarm is transient.
Over the past weeks it only occurred twice.
I just checked my T'Bolts and 2 of them displayed a partial PPS Adev
but a full OSC Adev table (lower). I just did a clear, C then a B.
I know not what caused that.
Could of been a utility power transient, LH/computer is on an UPS
but the T'bolt power supplies are not.
I am using a TC = 999 seconds with a dampening factor of 0.70
I am consistently getting ~ at 10K tau, low -13's
Stan, W1LE Cape Cod FN41sr
On 12/10/2012 6:37 PM, Sarah White wrote:
> On 12/10/2012 6:10 PM, Arthur Dent wrote:
>> I believe that the high temperature alarm you see is triggered at 50 degrees
>> C. If that is what you're seeing without artificially raising the temperature
>> of the Thunderbolt by insulating it so it can't radiate the heat, what I said
>> about replacing the chip is correct but if it is staying within a few degrees
>> over the course of the day and is in the 40 degree C range without being
>> insulated, the DS1620 thermometer chip is o.k..
> My particular thunderbolt seems to be from 2004, so I guess that manual
> is at or around checked:
> ThunderBolt[tm] GPS Disciplined Clock User Guide Version 5.0
> 5.0 is a version of the manual published in 2003
> Table C.2.2 Environmental Specifications:
> Operating Temp: -0°C to +60°C
> Storage Temp: -40°C to +85°C
> ... so I guess I shouldn't worry about a few degrees of difference in
> the reported temp, especially considering that the sensor in question
> isn't on the OCXO itself.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts