[time-nuts] 10 MHz output level from DATUM 9390...
ziggy9 at pumpkinbrook.com
Sat Dec 15 23:37:05 UTC 2012
On the scope I see 950mv RMS with 50 ohm termination, 1.5V RMS with 1M. This is on a 9390-6000 OCXO with the default timing output configuration, 10MHz on J7. Hope this helps.
Paul - K9MR
On Dec 15, 2012, at 4:21 PM, Burt I. Weiner wrote:
Does anyone have a DATUM 9390 series GPS receiver that can tell me what the (50 Ohm) terminated output level of the 10 MHz spigot is. One of mine started spurring and the other one is clean, but seems way to high in output level.
Burt I. Weiner Associates
Broadcast Technical Services
Glendale, California U.S.A.
biwa at att.net
time-nuts mailing list -- time-nuts at febo.com
To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
and follow the instructions there.
More information about the time-nuts