[time-nuts] NTGS50AA glitches
Chuck Forsberg WA7KGX N2469R
caf at omen.com
Tue Jun 12 11:31:03 EDT 2012
On 06/12/2012 07:20 AM, Dan Rae wrote:
> While generally behaving really well my NTGS50AA does apparently show
> an unusual sensitivity to being moved. The attached LH plot shows
> what happened when it was moved slightly on the bench two or three
> times. What surprises me is that the DAC plot remains offset after
> everything else has settled. It's not like I hit it with a hammer or
> anything. Is this normal behaviour?
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there
At this level of precision any bump is likely to cause the OCXO to
Is tere a way to update a Thunderbolt to the latest version of GPS?
Chuck Forsberg WA7KGX N2469R caf at omen.com www.omen.com
Developer of Industrial ZMODEM(Tm) for Embedded Applications
Omen Technology Inc "The High Reliability Software"
10255 NW Old Cornelius Pass Portland OR 97231 503-614-0430
More information about the time-nuts