[time-nuts] I can't get a nice waveform from my rubidium !
lists at rtty.us
Thu Apr 17 06:59:00 EDT 2014
I believe that you have a problem with that probe. A broken ground is one possibility, a significant mismatch between the probe output and the scope input is another possibility.
On Apr 17, 2014, at 1:32 AM, Claude Fender <labtf1 at yahoo.fr> wrote:
> Hello List,
> I've recently bought a Quartzlock 10A R (rubidium). Does anyone has experience with this unit?
> unit has two outputs with no options. The problem is the shape of the
> waveform (see attachments), it's the same shape for the two outputs.
> When I
> connect an output directly to the oscilloscope, the shape is nearly
> "squared" and when I use a probe, the shape is nearly "sine" !
> When I send a 10 MHz signal from a generator, the waveform is a clean sine with or whithout the probe.
> So what signal should I except from this rubidium ? Or most probably, what is wrong with my use of the oscilloscope ?
> (The frequency is ok by the way (+2E-10) and I didn't open the case yet)
> Thanks for any advice
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts