[time-nuts] How can one measure ADEV of a good oscillator?
kb8tq at n1k.org
Mon Dec 1 19:30:57 EST 2014
> On Dec 1, 2014, at 7:23 PM, Richard (Rick) Karlquist <richard at karlquist.com> wrote:
> On 12/1/2014 4:08 PM, Bob Camp wrote:
>> Others did a similar thing by simply taking production OCXO’s to the limit of their EFC range. That let you do a coarse sort to find some number of “likely” units. Next step was to pop a few of them open and short this or that out to get a reasonable beat note. Numbers in the 10 to 20 Hz range were pretty common. After that a single mixer setup followed by a simple limiter got things good enough to screen the production lots.
> I am fairly sure they would have done that if it was workable,
> to avoid special crystals. One rather huge problem with a
> 10 Hz beat note is that you are never going to measure ADEV
> at short (<1 second) sample times that way.
The technique was used for 1 second and longer ADEV screening. That’s what most of the specs were based on. Tight numbers at < 1 second were rarely seen. For what ever reason, phase noise stopped at 1 or 10 Hz and ADEV went from 1 second out to longer tau.
> Rick N6RK
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts