[time-nuts] LNA and Alias
magnus at rubidium.dyndns.org
Mon Dec 22 10:16:39 EST 2014
If you have a single-pole low-pass filter at say 20 kHz you will have a
60 dB damping as you hit 20 MHz. No reason to get really fancy.
Either just do a two-pole low-pass filter (inductance, capacitance) or a
two-stage RC-link, or consider doing a LC-series link between signal and
ground to swallow that 20 MHz, providing a notch.
Do it near the mixer, no need to polute the amplifier with the signal.
You can do another LC-link at the ADC end to make sure you swallow that
20 MHz. Consider adding additional such links for say 10 MHz or other
reasonable likely mixer products.
On 12/21/2014 09:39 PM, Loïc Moreau wrote:
> Hi all,
> My phase noise measurements system give erroneous results in close in phase noise measurements, I got humps in the 1Hz-1000Hz area as high à 20 dB more than expected.
> The setup is using a mixer to compare reference and DUT witch drive an LNA http://www.ko4bb.com/~bruce/LowNoiseMixerPreamp.html , the output is driving an AD7760 ADC and an op amp circuity is connected to the VFC reference to achieve quadrature. The results are analyzed with an homebrew FFT charting software
> After struggling with different configurations, switched different LNA, ADC , sound card. A scope connected to the LNA output indicate steady 20MHz residuals just before the ADC ( around 10 mv peak-peak).
> In fact , it seems that the mixer 20Mhz residuals ( DUT + REF ) are entering the ADC and so theses alias give erroneous results in the 1Hz-100Hz area, displaying unexpected artifacts. In order to fix the problem I will probably include an analog filter just before the ADC input (same as LNA input 1nF 80µH), but I want to know if some more sophisticated measures should be undertaken as an 5th Order Lowpass Filter.
> I have not found many clue about alias problems in phase noise measurements literature so I may have missed something ?
> Any advice
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts