[time-nuts] FASTRAX GPS
jwsmail at jwsss.com
Tue Jul 1 17:56:13 EDT 2014
I went back thru this thread to look for the answer, and did not see it
so am asking. The one of these I have has a threaded female connector
on it. One of the ones in Tom's page has a smooth sided connector, FWIW.
I wonder if this would work with one of the 5m SMA male active antenna
units will work with it?
This one happens to be 5 meters long and is active. I am guessing
because the unit has antenna power this is the sort of item to use?
I want a long run because where I plan to use it is some distance from
clear line of site.
On 6/20/2014 8:31 AM, Tom Van Baak wrote:
> Hi Ernie, Jason, (also Hal, Chris),
> I'm able to get NMEA and 1PPS out of the Fastrax/iTrax130 board now.
> Before I sink any more time into this project, have any of you made 1PPS measurements?
> Compared to ublox, using the same antenna, these units take tens of minutes or even hours to lock. And they have a bad habit of being close to UTC for a few hours but then gradually wandering off by tens or hundreds of microseconds.
> A power cycle puts them back on track (aligned with UTC), so it would seem to be a firmware issue rather than antenna or reception. A similar thing happens on three different boards I have evaluated over the past two weeks.
> Now, when they are working right, the 1PPS has an RMS deviation of around 20 ns, which I've come to expect as typical for cheap GPS/1PPS receivers these days. But the long lock times and unexplained 1PPS drift make them unreliable for GPSDO or serious timing work, even at $12 each.
> It might be me, so I'm asking if you've seen anything similar. The binary command set looks tempting, but I usually don't play with that until the unit can be trusted to give reliable NMEA and 1PPS output.
> Unfinished page: http://leapsecond.com/pages/itrax/
> ----- Original Message -----
> From: "Ernie Peres" <ernieperes at aol.com>
> To: <time-nuts at febo.com>
> Sent: Wednesday, June 11, 2014 4:11 AM
> Subject: Re: [time-nuts] FASTRAX GPS
>> Hi Jason,
>> I figured out a different pin assignments....
>> hold the PCB so that the SMA antenna connector is looking to right and under the PCB.
>> open- 1 2- Ant power presently connected to pin nbr 4
>> TX- 3 4- Main power +3 volt
>> RX- 5 6- 1PPS
>> open- 7 8- Grnd power -3 Volt
>> the unit takes about 60mA @ 3,0 volt and comes-up in NMEA mode.. 9600 Baud.
>> I use the FASTRAX_WORKBENCH_522 software.
>> I wonder if anybody has other pin arrangement/connection. The 2 open pins is UNK and also not sure if pin nbr 4 is the main power or perhaps the pin nbr 2......??
>> Rgds Ernie.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts