[time-nuts] Another "atomic" clock question
lists at rtty.us
Sun Mar 2 19:50:41 EST 2014
I believe the paper was by Stan Shadowski. I’m *certain* I’ve mis-spelled his last name, which is indeed a very poor move on my part. I would not be surprised if there are several co-authors.
I don’t have the UFC indexes here at home so I have no quick way to look it up.
On Mar 2, 2014, at 7:40 PM, Jim Lux <jimlux at earthlink.net> wrote:
>>>> As Jim mentions in another post, you can run on the fundamental and the third (or 5th or 7th) and get a thermometer out of the delta between the two modes. The gotcha is that a change in load impedance will shift the frequencies unequally. That will give you an apparent temperature change.
>>> I already know about the fundamental and third trick, my question was if it could be done to AT-cut as well. I interpret your statement as yes, it does. I don't trust it to be perfect, but reasonable. Ideas for means to handle shift would be welcome.
>> It was originally proposed by a very nice guy from Ft. Monmouth for use with AT cut resonators. I believe the paper is in the FCS proceedings from the mid 1980’s. The DOD kept rights to the technique and licensed it to a couple of oscillator companies.
> Hmm. SC cut, perhaps? (see the third reference down..
> R. L. Filler and J. R. Vig, “Resonators for the microcomputer compensated crystal oscillator,” 43rd Ann. Symp. Freq. Contr., pp. 8- 15, 1989.
> there's also
> The microcomputer compensated crystal oscillator (MCXO)
> Bloch, M. ; Frequency Electron. Inc., Mitchel Field, NY, USA ; Meirs, M. ; Ho, J.
> The MCXO uses a novel technique to achieve temperature compensation without the use of ovens or conventional temperature-compensating components. The crystal oscillator in the MCXO, which is free to vary with temperature, operates on two modes simultaneously-the fundamental and the third overtone. Several advantages accrue because this method of temperature compensation does not resort to frequency pulling. The authors presents the details of how the MCXO operates and the details of the performance of the delivered systems
> Published in:
> Frequency Control, 1989., Proceedings of the 43rd Annual Symposium on
> Date of Conference:
> 31 May-2 Jun 1989
> 16 - 19
> Meeting Date :
> 31 May 1989-02 Jun 1989
> INSPEC Accession Number:
> Conference Location :
> Denver, CO
> Digital Object Identifier :
> But then,
> Yoonkee Kim (from Ft Monmouth)
> has a paper (DTIC ADA484423)
> Aging of Dual Mode Resonator for Microcomputer Compensated Crystal Oscillator
> Abstract— A Microcomputer Compensated Crystal Oscillator (MCXO) utilizes the dual c-mode excitation (fundamental mode and 3rd overtone (OT)) of an SC-cut resonator for self- temperature sensing and compensation. The long-term stability of the MCXO depends primarily on the aging of the dual mode resonator. When two modes age differently in time, the aging MCXO’s output frequency curve would shift with a tilt over its operating temperature range
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts