[time-nuts] Lucent KS-24361, HP/Symmetricom Z3809A, Z3810A, Z3811A, Z3812...
goetz at g-romahn.de
Sun Nov 2 04:18:50 EST 2014
> So that's it folks, after all this it would now seem that all that's
> needed to enable a Ref-1 unit stand alone is to link together J5 pins 2, 10, 12,
> and 15, and to ground pin 3 to pin 8, and then just hang around for hours
> and hours on end with yer fingers crossed:-)
> I don't know if it's possible to monitor the J8 Diagnostic port whilst
> waiting for it come live, that would at least give a better idea of what's
> going on, but I deliberately avoided any other connections whilst trying this
> out so it's possible that might be inhibited also.
> Anyway, apologies for all the waffle, I've been topping this up as I go
> along whilst surviving on coffee and getting a bit groggy in the process,
> it's 0730 here now and no sleep yet, but once I'd had my first encouraging
> results there was no way I could just go to bed and leave it:-)
I connected pin 4 also to ground permantly without observing any
negative sideeffects. SDo no active electrics needed but you will (want
?!) not see STBY led blinking.
More information about the time-nuts