[time-nuts] Lucent KS-24361, HP/Symmetricom Z3809A, Z3810A, Z3811A, Z381...
kb8tq at n1k.org
Mon Nov 3 18:55:49 EST 2014
I got a chance to look at phase noise today. I’ll try to post the plots later. Quick summary:
10 MHz - ugly. Lots of spurs, many of them close in. Poor phase noise floor.
15 MHz - pretty good. Noise floor is not as good as a TBolt (by 5 to 15 db). Far fewer spurs than the 10 MHz, fewer spurs than the TBolts I’ve looked at.
ADEV on a short run looks better than an un-tuned TBolt. I’ll need a much longer run to see what it’s really doing.
> On Nov 3, 2014, at 1:20 PM, GandalfG8--- via time-nuts <time-nuts at febo.com> wrote:
> Ah, that's interesting, perhaps it wasn't me after all then.
> Did you have the same flashing ON light symptoms?
> In a message dated 03/11/2014 17:56:56 GMT Standard Time,
> hmurray at megapathdsl.net writes:
>> It turns out this is what happens if you switch the "Output Level" from
>> to 23, obviously an advisory indication to draw attention to the higher
>> output. Switching it back reduces the level, as expected, and returns the
>> LED function to normal. Phew:-)
>> I can't remember switching it but don't suppose it arrived like that so
>> guess I must have done.
> My pair came with Ref-0 set at 17 and Ref-1 at 23.
> These are my opinions. I hate spam.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts