[time-nuts] My NTGS50AA failed
eb4apl at cembreros.jazztel.es
Wed Nov 12 19:33:10 EST 2014
Thank you, I found this very enlightening. Now I have to find the 9.7
KHz pulses, no luck so far.
El 12/11/2014 a las 7:34, Charles Steinmetz escribió:
> Ignacio wrote:
>> I'm not very sure that the DAC is working, I suppose that the unit
>> doesn't measure the DAC output, it reports the DAC commands. My
>> voltage figures is what LH reports (so the NTGS50AA reports, probably
>> what it is trying to do), but the frequency control pin of the
>> oscillator is stuck at 5.02 V regardless of the supposed (intended)
>> DAC output, it does not move at all.
> One more thing that could be helpful -- on Nov. 2, 2013 Stewart Cobb
> posted a description of the DAC operation for a Trimble Thunderbolt
> ("Thunderbolt tuning DAC theory of operation"). On Jan. 5, 2014
> someone (no name) posted further information on the same thread re:
> the Trimble/Nortel 45k. You can find both posts in the list archives
> (follow the link at the bottom of this post, then click "time-nuts
> Archives" on the list home page).
> Best regards,
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the time-nuts