[time-nuts] ***SPAM*** 53230A input channel fault?
michael.cook at sfr.fr
Wed Nov 19 05:07:35 EST 2014
> Le 19 nov. 2014 à 09:47, Anders Wallin <anders.e.e.wallin at gmail.com> a écrit :
> Hi all,
> I have a 53230A that behaves differently on ch1 and ch2.
> With ch2 at DC/1Mohm/postive-slope/1V it triggers ok on my PPS signal and
> shows e.g. a pusle-width measurement etc.
> The same signal connected to ch1 fails to trigger.
Are you sure you set the level to +1V…. That got me once. ;-)
> I noticed the Vpp and Vmax/Vmin voltage readings are not correct for PPS
> signals - i.e. on a scope my signals are from 0V to 4V, but the counter
> shows -5 mV to +5 mV or similar. But perhaps that is because of the low
> duty-cycle of a PPS signal.
Mine shows the same and I assumed that this is the case, though I haven’t delved further.
> We were measuring RF frequencies with the counter a few weeks back - it
> says max 1W input... it is possible that this was exceeded (does 50Ohm vs.
> 1MOhm setting matter for input-protection?).
> Any ideas on what breaks first on the input stage? :) (are there schematics
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts