[time-nuts] Measurement of frequency of HP 8720D option 1D5 oscillator after switch on
magnus at rubidium.dyndns.org
Thu Oct 2 02:10:26 EDT 2014
The character of starting high/low and then stabilize some 5-30 min
later is typical of oven oscillators. Underdamped ovens have been seen
before, I have even seen one on the brink of oscillation.
TCXO will not have the same wide range, as it compensate for the
On 10/01/2014 10:11 PM, Dr. David Kirkby (Kirkby Microwave Ltd) wrote:
> Further to my question the other day about what type of oscillator was
> used in the HP 8720D VNA, fitted with the high stability oscillator
> option (1D5), here is the frequency as the instrument is switched on,
> after being powered off for 2-3 hours.
> The oscillator appears to start too high in frequency, overshoots
> downwards, then settles down. It would be a somewhat useless exercise
> collecting data for much loonger and looking at in detail, since I
> don't know the accuracy of the HP spectrum analyzer. But that had been
> on for at least 24 hours.
> The measurement system was an HP 70000 system spectrum analyzer,
> fitted with the 70310A precision frequency reference. This has not
> been calibrated in years.
> I will get one of those Jackson labs GPS frequency references, then I
> will know what the frequency is!!
> Assuming the crystal frequency is only a function of temperature, (and
> I know that not to be the case), it looks as if the oven is under
> damped to me.
> Is this behavior typical of a TCXO, OCXO or whatever ??
> Any comments???
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts