[time-nuts] Efratom FRT-H Lamp removal and supply schematic
sebas.diaz at gmail.com
Sat Oct 25 12:44:20 EDT 2014
Followed your advice, and made a 15x2mm screwdriver by grinding down a
carbon steel scraper. A combination of 10min heating on the physical
package, plus the tool and got the rubidium bulb out.
I've had these oscillators running since 8:00 in the morning, doing a
simple XY signal tracing comparing both. The older unit loses lock
semi-periodically, and then relocks after a random time from 4 to 10
minutes. The newer unit hasn't lost lock, but there may be an issue with
the sma connector.
On Fri, Oct 24, 2014 at 11:07 AM, <cdelect at juno.com> wrote:
> Let the unit warm up for 10 minutes, then power down and then try the
> lamp removal.
> It turns CCW to remove.
> Use a BIG screwdriver!
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the time-nuts