[time-nuts] What sort of oscillator is this?
saidjack at aol.com
Sun Sep 28 13:40:32 EDT 2014
An ocxo has two effects that cause a frequency change after power on: heater stabilization and crystal retrace.
Heaters usually stabilize quickly (1 - 2 minutes for DIP-14 ocxo, 7 to 10 minutes for typical eurocan docxo's) and then a. ~30 minutes soak until the ocxo starts following ambient temperature.
After the initial warmup the crystal goes into an exponential retrace as it out-gasses etc and it can take many hours or even days or weeks for that to subside. Then what is left is typically linear aging.
Sent From iPhone
> On Sep 28, 2014, at 3:16, Hal Murray <hmurray at megapathdsl.net> wrote:
> drkirkby at kirkbymicrowave.co.uk said:
>> Two people responded - one says a OCXO and the other an TCXO!!
>> The warmup time is I think an hour, but clearly that is not the time for an
>> oven to warm up.
> An hour seems like a reasonable OCXO warm-up time to me. You might get
> faster warm-up times, but you will probably pay for it someplace else. Here
> is a graph:
> Also, it probably depends upon how long it has been powered off.
> See how much power the system draws when plugged in but turned off. Most
> gear using OCXOs keep them powered up when the front panel switch says "off".
> So 0 is a strong indication that you have a TCXO. You can also unplug it
> for a couple of hours, then watch the power after you plug it in. I'd expect
> a step decrease in power after the core is mostly warm.
>> I find it odd that an instrument that probably cost $50,000 when new did not
>> have a TCXO as standard, and perhaps an oven as an option.
> Maybe most customers have a good lab source of 10 MHz and use an external
> These are my opinions. I hate spam.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts