[time-nuts] Bad date out of HP Z3805A
hmurray at megapathdsl.net
Sun Nov 1 21:20:28 EST 2015
charlie at drhabekost.com said:
> I did a system reset, and a new survey. The date on startup is 1/1/1996; It
> jumps to 3/17/1996 and stays there (over a week now). All other functions
> appear to be normal. Power off/on, no difference.
> Am I missing something?
That sounds like the 1024 week rollover problem.
I don't have a 3805, but I had similar troubles with a 3801.
The 3801 has a command to set the date. You can find the details in the
manual which is readily available. You may have to set the date before it
gets the date from a satellite. That may be easier if you disconnect the
antenna and then power cycle.
These are my opinions. I hate spam.
More information about the time-nuts