[time-nuts] Bad date out of HP Z3805A
Graham / KE9H
ke9h.graham at gmail.com
Sun Nov 1 22:35:36 EST 2015
Yes, you are missing 1024 weeks.
Google: GPS week rollover.
On Sun, Nov 1, 2015 at 7:19 PM, Charlie <charlie at drhabekost.com> wrote:
> I have used a HP Z3805A for my shack for a number of years. I recently
> wished to get the date output. Hooked up my laptop, and found that while I
> had a stable, in spec 10 Mhz output, I now have a date of 3/17/1996.
> I did a system reset, and a new survey. The date on startup is 1/1/1996; It
> jumps to 3/17/1996 and stays there (over a week now). All other functions
> appear to be normal. Power off/on, no difference.
> Am I missing something?
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the time-nuts