[time-nuts] PRS10 1 PPS Out having random phase jumps
mark at alignedsolutions.com
Mon Nov 23 16:46:29 EST 2015
I don't recall ever having this issue with mine. As a general rule though I seemed to have more success using 5 and 10 MHz reference signals vs 1 pps reference signals with my HP5370's and HP5335's when making Adev measurements of 5 and 10 MHz OCXO's.
Sent from my iPhone
> On Nov 23, 2015, at 12:11 PM, James Robbins <jsrobbins at earthlink.net> wrote:
> Has anyone using the 1 PPS output from a PRS10 rubidium found significant
> random phase jumps? I'm getting these random phase jumps while using the
> PRS10 as a TI base for ADEV. I can use the 10 MHz output of the PRS10 to
> feed a T2mini without any similar jumps. Wondering if it might be heat
> related or other. Needless to say, such jumps screw up the TI ADEV readings
> and convert the TimeLab readings from the E-12 range to E-1 or so. Thanks.
> Jim Robbins
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts