[time-nuts] ACAM GP22 Chip
th.allgeier at gmail.com
Tue Nov 24 09:04:32 EST 2015
I have an ACAM GP22 TDC chip and evaluation board which I am looking at for “work” purposes – I work for a company active in the weighing and force measurement world.
I should say from the start that I am new to time and frequency measurements and not even an electronics engineer – but then I have been exposed to high-precision electronics for the last 25 years hence have picked up some dangerous degree of half-knowledge.
We want to use this chip to measure the period of a square wave, of around 13 kHz i.e. in the 70 µs range. As the application is potentially high-accuracy we need to know the period to within 1 ns or better.
In order to evaluate the chip I was planning to replicate John A’s experiment with the coaxial delay line from the HP5370b – but as my interest is in “measuring range 2” of the GP22 I need a delay of 500 ns or more (actually 1 µs sounds a better start). This is the equivalent of a 200 m length of cable. I fear trouble with this: Am I not getting unwanted inductivities if I use a coil of that size?
So, to come to the point: Am I pushing the concept of a coax delay too far with 1 µs and are there other (simple/reliable) ways to achieve this kind of delay? I have tried it with a shorter piece of cable (around 2 ns which is measured in “range 1”), there I seem to get consistency virtually to within 100 ps. But I need to know if the device sticks to this level of performance when the periods are much longer, and thus measured in “range 2”.
Thanks and best regards,
More information about the time-nuts