[time-nuts] State of the art of crystal oscillator measurements
kb8tq at n1k.org
Thu Aug 11 17:50:05 EDT 2016
The gotcha is that you don’t *have* to have a 50 ohm system. An output stage with
a narrow band tuned tank is one example of a very “not 50 ohms” system.
There is also a whole debate around the “is 50 ohm source into 50 ohm load really 25 ohms”.
That will give you a 3 db delta to bet beers about. The 3 db split of thermal noise between AM
noise and PM noise is fairly well accepted, but it still can be challenged.
So is the floor -174 - 3 = -177 or is it something else ….
If we have a -195.4 dbc/ Hz oscillator that puts out < 18 dbm in a 50 ohm system it’s not -177 or
it’s not a 50 ohm system.
> On Aug 11, 2016, at 4:47 PM, John Miles <john at miles.io> wrote:
> Or rather -(177+DUT output power in dBm). The minus sign makes the difference between the thermal floor and a nuclear war!
> -- john, KE5FX
> Miles Design LLC
>> Remember that L(f) is expressed in dBc/Hz, not dBm/Hz. If it were dBm/Hz,
>> then kT would be the limit. But in dBc/Hz terms, the limit is 177 + the DUT's
>> output power in dBm.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts