[time-nuts] HP53131A RS232 question
kb8tq at n1k.org
Fri Dec 2 15:15:34 EST 2016
I don’t typically use 50 ohms with logic signals, but other than that I
don’t see much different between what you describe and what “just works”
for me on multiple 53131’s and 53132’s. About the only weird thing I can think
of would be a baud rate so slow that the string does not complete. I
don’t think that the counters will go that slow …..
I never use any form of hardware or software pacing, A “send after timeout”
might explain some weird stuff.
> On Dec 2, 2016, at 2:09 PM, cdelect at juno.com wrote:
> I reconfigured for TI with +.7V level 50Ohm and I get a good count.
> With a TI of around 43milliSec between two 1pps signals the display is
> updating every second.
> However if I turn print on the display update slows way down and no
> longer displays consecutive counts!
> I know I had this working once!!!
> Should I be using some other function than print?
> I just want the hyperterminal to take and log the data!
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts