[time-nuts] Temperature weirdness with Thunderbolt & Lady Heather 5
holrum at hotmail.com
Thu Dec 15 12:45:40 EST 2016
Version 5 can write a "raw" receiver data capture log with the WY keyboard command.
Heather has code for "un-doing" the temperature spikes from the Tbolt temperature sensor/firmware bugs. The "/tj" command line option control attempts to remove the spikes. "/tj" toggles the spike filter "/tj0" forces it off "/tj1" forces it on. Earlier versions of Heather defaulted the spike filter to ON, the current version defaults it to OFF since not all Trimble compatible devices or Thunderbolts behave the same and enabling the filter on devices without the spike issue can mess things up.
More information about the time-nuts