[time-nuts] HP 106a oscillator connectors question
rdarlington at gmail.com
Fri Jun 3 21:29:23 EDT 2016
They look like microdot connectors. Like what I have on my ultrasound
On Jun 3, 2016 3:59 PM, <cdelect at juno.com> wrote:
> Well the checkout of the 106A oscillator section is going well.
> I ran it for 6 hours and the outer oven came down OK, will have to wait
> until I can do a longer attended run for the inner oven as it takes 8 1/2
> hours per the manual to come down!
> This unit was not built with the fine adjust via an internal pot as shown
> in the manual but does have the external EFC jack show. (EFC jack does
> After the 6 hours Frequency was within a HZ and could be adjusted. Quit
> there until I can find a spot for it to live for an extended test.
> Need help identifying the RF connectors on the unit.
> They are not SMA but look to be the same diameter. would need two
> adaptors or a couple old cables with the matching connector I could
> splice into. I had to poke in a wire and use clip leads!
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the time-nuts