[time-nuts] And another Thunderbolt question
michael.cook at sfr.fr
Tue Aug 8 13:09:26 EDT 2017
> Le 8 août 2017 à 17:53, Vlad <time at patoka.org> a écrit :
> Hello Netizens,
> After the power outage, my TB start to shows some strange behavior.
Was it a normal power outage , or lightning strike ? Mine just went through a normal power outage without issue.
> As I start LH, the screen shows like this (You could see the date stamp as Dec 1997_
This is normal now that the GPS week rollover has occurred for these devices.
Strange that there are no sat stats as the antenna status looks ok, there is an almanac and it would appear that the date and time are being retrieved for some sat .
> Then, few seconds after, the date will be corrected
This is normal . The wizard has corrected the bad data.
> And after another few seconds, I have a almost dark screen like this
I haven’t seen this
> I tried to use ThunderBolt Monitor software to reset my TB to factory settings. It didn't helps. This monitor always shows me correct time, but date back from 1997. And its always no sats.
Tboltmon has no date fix. Again strange that no sat data is available, like the receiver is not reporting sv data to the application.
> What else could be done to resurrect the device ? Thanks !
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
"The power of accurate observation is commonly called cynicism by those who have not got it. »
George Bernard Shaw
More information about the time-nuts